catheterization task trainer
1. Assembly and disassembly
Recommended equipment sizes
16FR 5.3mm catheter
Setup and maintenance
Make sure that the red valve screw is tightly screwed to the base.
Fill the base with water. The water should almost completely fill the container and cover the horizontal bar.
After swapping the genital inserts, double-check if the red pin sits properly inside the silicone.
If the genital inserts tend to leak before the catheter is inserted, unscrew the red valve screw and check the silicone valve. Add a little bit of lubricant inside it and re-attach it to the base.
Always lubricate the equipment prior to use. Silicone grease and water-based lubricants (like optilube) work well.
Store in clean, dry conditions away from heat and direct sunlight; avoid contact with metals, solvents, oils or greases, and strong detergents. When the product is not in use please store it in the carrier case provided.